DP Level Measurement ExplainedLearn about open vessel level measurement using a differential pressure transmitter.
In this first article of a 2-part series, we’re going to introduce you to open vessel DP level measurement using a differential pressure transmitter.
The second part of our 2-part series will take you through the process of predicting Differential Pressure Transmitter outputs of open and closed tank level control loops.
Even with the development of new technologies and highly sophisticated level measurement instruments, Differential Pressure Transmitters are still in service today measuring vessel fluid levels.
DP level measurement
In this article, we’re going to:
– Discuss the two categories of level measurement
– Talk about the difference between Direct and Indirect methods of continuous vessel level measurement
– Explore how a Differential Pressure Transmitter is used to measure vessel level and discuss variables that influence the measurements
Ok, …off we go!
2 categories of level measurement
Level measurement can be divided into two categories: point and continuous.
1) Point level measurement
Point level sensors are used to mark a specific single liquid height.
For example, the point level sensor would send a signal to a PLC at a specific level to start a pump.
2) Continuous level measurement
A continuous level sensor measures the fluid level within a measurement range rather than at a specific single point. Said a different way, continuous level sensors provide level measurement over the full 0% to 100% range of a vessel.
Because we are interested in the height of a liquid in a vessel, we take vertical measurements taken from the surface to a fixed reference point. The reference point is usually near the bottom of the vessel.
You might have noticed that the vessel Reference or Zero percent point is not at the very bottom of the vessel.
This is to make sure that the vessel is never completely drained.
We can take advantage of the fact that the pressure measured at the base of a liquid storage vessel will vary with the height, or level, of the liquid in the vessel.
Armed with that knowledge, we can determine the level of the vessel from the indirect inferred measurement of the pressure.
Level measurement factors
We don’t have to worry about the size, shape, or volume of the tank because none of those parameters have any effect on the pressure at the reference point.
But, we do have to consider other parameters such as specific gravity and temperature.
DP Level measurement
Alright, we’re ready to look at how a Differential Pressure Transmitter can be used to measure the vessel level.
We said earlier that the pressure at the bottom of a vessel is directly related to the level of the liquid in the vessel. In that case, if we connect a Differential Pressure Transmitter to the reference, or zero percent point of that vessel, we can use the pressure measurement to determine the level.
Another variable that has a major effect on the pressure at the reference point of the vessel is the relative density of the liquid in that vessel. Relative density is also called specific gravity.
There is a relationship between pressure, relative density, and liquid height. That relationship can be expressed as:
Pressure (inches of water) = Relative density × Liquid height (inches)
Now… let’s see how we can measure the level of an open vessel using a differential pressure transmitter.
The High-Pressure Port is connected at the 0 meters point and the Low-Pressure Port is vented to atmosphere.
How can we predict the pressure at the bottom of the vessel?
Consider our discussion earlier about the relationship between liquid height, relative density, and pressure in inches of water to determine the pressure at the bottom of the vessel for any liquid level.
DP Level measurement – pressure scale
You can convert the resulting inches of water pressure value into any pressure scale you need such as psi, kPa, bar, etc.
Let’s review what we’ve discussed.
– Level measurement can be divided into two categories: point and continuous.
1) Point level sensors are used to mark a specific single liquid height.
2) A continuous level sensor measures the fluid level within a measurement range rather than at a specific single point.
– Vessel Level can be determined indirectly using a Differential Pressure Transmitter by measuring the pressure at the 0% point of the vessel and inferring the level.
– For vessel measurements, Pressure in inches of water is equal to the relative density of the liquid multiplied by the height of the surface of the liquid in inches.
Want to Learn More?
Newcomers to RealPars have free access to one of our online courses.
You can take advantage of this offer by downloading the RealPars app (iOS, Apple TV, Android, Android TV, FireTV, Roku) and selecting the first lesson of the PLC Hardware course. Just like the video, the lessons are all high-quality and also very easy-to-follow.
Also, for a low monthly fee, you’ll get full access to an exclusive library of courses on PLC programming and industrial automation topics as well as new fresh out of the oven videos each and every week. These courses are all member-exclusive and are not shared anywhere else on the internet.
If you would like to get additional training on a similar subject please let us know in the comment section.
Check back with us soon for more automation control topics.
Got a friend, client, or colleague who could use some of this information? Please share this article.
In this blog post, you’ll learn about the mindset that helped me getting a PLC programming job with NO experience. This is my personal experience as someone who searched for a job in this field and as an employer who reviews resumes and interviews candidates for a variety of projects. So let’s get started!
In this easy-to-follow article, we're going to talk about a very important part of any automated industrial process the Safety PLC.Safety Instrumented System (SIS)A Safety PLC is one of the 3 parts of a Safety Instrumented System (SIS). The SIS can also be referred to...
In this article, we're going to explain how to read analog input control loop diagrams.Industrial control loopsEvery industrial control feedback system has 2 loops: an input control loop and an output control loop.Analog input loopThe analog input loop consists of 2...
Learn how to program PLCs, install and wire industrial devices, and at the same time purchase them online.
+31 10 316 6400
Mon - Fri 8:30 am to 5:30 pm (CET)
Rotterdam Science Tower, Marconistraat 16,
3029AK Rotterdam, Netherlands
© 2020 RealPars B.V. All rights reserved.
Created with coffee and tea in Rotterdam.