How to measure Flow Rate with a DP Transmitter?Learn how a Differential Pressure Transmitter is used to measure the volumetric flow rate.
In this article, we’re going to introduce you to flow rate measurement using a Differential Pressure Transmitter.
Before we get started, you might want to review two of our other videos:
Flow transmitter calibration is the focus of our discussion, but let’s review some other important terms first. We’re going to define the terms Range and Span and Zero and Span adjustments.
After that we are going to:
– Explain how a Differential Pressure Transmitter is used to measure the volumetric flow rate
– Explain why a Square Root Extractor is needed in Differential Pressure Transmitter flow measurements.
Ok…let’s go! We say that an instrument is calibrated when its output corresponds to a specific input across a specific range of inputs.
Said another way, Instrument calibration is a process where the instrument is adjusted in order to achieve the desired output range for a specific input range.
For example, a current output type Flow Transmitter is calibrated for a volumetric flow range of 30 gallons per minute to 100 gallons per minute.
When the transmitter produces a current output of 4 milliamps at 30 gallons per minute and a current output of 20 milliamps at 100 gallons per minute a flow rate of 65 gallons per minute would produce an output of 12 milliamps.
Range and Span, LRV, and URV
The input Span of an instrument is simply its Lower Range Value often referred to as LRV, subtracted from its Upper Range Value, often referred to as URV.
For example, the Span of the input range of 30 gallons per minute to 100 gallons per minute is 70 gallons per minute.
Zero and Span
Zero and Span are the names of the adjustments made in order to calibrate an instrument. There are many ways to adjust zero and span.
For example, they could be physical adjustments such as potentiometers, or software-based push-button adjustments.
Flow Rate Measurement
A common method of flow measurement is done by using a Differential Pressure Transmitter. The Differential Pressure Transmitter often referred to as a Delta P transmitter, is placed across an obstruction such as an orifice plate.
The orifice plate will cause a varying differential pressure drop across it as the flow through the pipe changes.
Square Root Extractor
Unfortunately, the Differential pressure across the orifice is not proportional to the flow rate but is actually proportional to the square of the flow rate. That’s why in applications like this, we need a Square Root Extractor.
Sometimes this square root function is built into the transmitter and sometimes a Square Root Extractor is a separate signal conditioning instrument connected to the output of the transmitter.
Flow Rate Measurement Calculation
Do you remember when we said that the differential pressure is proportional to the square of the flow rate?
In fact, the volumetric flow rate is directly proportional to the square root of the differential pressure.
You might have seen this equation before where Q is flow rate and Delta P is the differential pressure across the orifice.
Let’s do some math and see what happens. According to the equation, a flow rate of 90% would produce a differential pressure of 81%.
There is a way for the flow rate to be directly proportional to the differential pressure, and that’s by removing the square root from the equation.
A Square Root Extractor will perform this function. If you are still a bit fuzzy, it will all become clearer as we progress to Part 2 of this article series.
Let’s review what we’ve discussed.
– An instrument is calibrated when its output corresponds to a specific input across a specific range of inputs.
– The input Span of an instrument is simply its Lower Range Value subtracted from its Upper Range Value.
– Zero and Span are the names of the adjustments made in order to calibrate an instrument.
– A common method of flow measurement is done by using a Differential Pressure Transmitter across an obstruction such as an orifice plate.
– The differential pressure across an orifice is proportional to the square of the volumetric flow rate, therefore, we need a Square Root Extractor.
Want to Learn More?
Newcomers to RealPars have free access to one of our online courses.
You can take advantage of this offer by downloading the RealPars app (iOS, Apple TV, Android, Android TV, FireTV, Roku) and selecting the first lesson of the PLC Hardware course. Just like the video, the lessons are all high-quality and also very easy-to-follow.
Also, for a low monthly fee, you’ll get full access to an exclusive library of courses on PLC programming and industrial automation topics as well as new fresh out of the oven videos each and every week. These courses are all member-exclusive and are not shared anywhere else on the internet.
If you would like to get additional training on a similar subject please let us know in the comment section.
Check back with us soon for more automation control topics.
Got a friend, client, or colleague who could use some of this information? Please share this article.
In this blog post, you’ll learn about the mindset that helped me getting a PLC programming job with NO experience. This is my personal experience as someone who searched for a job in this field and as an employer who reviews resumes and interviews candidates for a variety of projects. So let’s get started!
In this first article of a 2-part series, we’re going to introduce you to open vessel DP level measurement using a differential pressure transmitter. The second part of our 2-part series will take you through the process of predicting Differential Pressure Transmitter...
In this article, we’re going to take a close look at the 2-wire transmitter and 4-wire transmitter and discuss where they are used and why. We’re also going explore transmitter wiring configurations and how transmitters are connected to a PLC. Before we get started,...
Learn how to program PLCs, install and wire industrial devices, and at the same time purchase them online.
+31 10 316 6400
Mon - Fri 8:30 am to 5:30 pm (CET)
Rotterdam Science Tower, Marconistraat 16,
3029AK Rotterdam, Netherlands
© 2020 RealPras B.V. All rights reserved.
Created with coffee and tea in Rotterdam.